Either way, I'm putting the description of it online here in case other people are interested. Note that 'we' designed it based on what we found available from various websites, so the design is hardly unique.
Also note that it requires a multimeter/voltmeter to get a reading -- there's no USB/serial output or anything fancy like that.
(in related news I'm planning on taking night classes in electronics at a local TAFE (which is like a community college) so that I can start building more instruments/toys myself. A potentiostat would be nice...)
Anyway, back when we built this we ordered the parts from Digikey (I lived in the US at the time), and here's a list of what we got as shown on the invoice (the parts numbers of the linked items differ in some cases -- I presume that they are the same, but cannot guarantee that they are.
* Note that you'll need more than one of some items.
* Note that the parts includes stuff for using a whiteboard/prototyping board i.e. we first made a non-soldiered version, and then made a permanent assembly.
* The stuff for the final pH meter (i.e. the stuff in the pomona box) is shown in red
* The stuff for connection the pH meter to a voltmeter/multimeter is shown in blue
* The stuff that's needed for the power supply is in bold black
* Stuff that I ordered at the same time but can't remember what we used it for is in the default colour (i.e. black)
Price Item number Description
--------- --------------------- --------------------
3.05 LMC6081IN-ND IC OP AMP PREC CMOS SINGLE 8-DIP
2.96 ACX1046-ND Conn. enc. bulkhead female jack x 2
3.78 7-1437529-5-ND Conn socket dip 8 pos gold T/h
7.71 J6212-ND BNC cable
8.07 501-1032-NB BNC female-dbl banana
47.85 945-1081-ND Converter AC/DC 15W +/-12V out DL T/H
In addition, looking at the board (see pictures below), you'll also need
3 x female banana sockets
2 x K5M104(?) capacitor (x 2)
2 x BC1020TR-ND BC102 capacitor (x 2)
3 x female banana sockets
and lots of wires.
You'll measure the potential (emf) using a voltmeter, and by calibrating the potential against a set of pH standards you can calculate the pH. In theory 0.0 mV should be at pH 7, and the potential should increase ca 59 mV per pH unit, so that pH 6 is +59 mV and pH 8 is -59 mV.
Whether this is actually true (it won't be) depends on your pH electrode, temperature, ionic strength etc.
|The BNC cable is connected to a double banana contact, which attaches to the voltmeter|
You can get a proper one, or build one yourself. I did the latter. It ain't pretty -- in fact, the following pictures should horrify you. NOTE: unless you know what you are doing you MIGHT DIE! Playing with high voltage stuff may also be ILLEGAL for unlicensed people in some jurisdictions.
|Power OUT -- +12 and -12 V, and ground.|